Message 5 of 5. So where can I set resolution to like 1Hz so that I can take more than 2 measurements in a second. Hi After update my A from firmware 1. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I finally found the manual and tried some We are having overhead issues with the new A counter that need to be resolved. Hi, I’d appreciate any help in reading data from an HP B.
|Date Added:||13 February 2015|
|File Size:||70.45 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Log in to follow, share, and participate in this community. We have recently added a reset step to our tests that use the a counter and every so often the counter returns a measurement interpolator calibration failed error during the subsequent setup of the counter. We are having overhead issues with the new A counter that need to be resolved.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I managed to solve the issue.
Hi, I’m very green in this area, so please excuse my dumb question. Anyone can help me how to communicate through RS Looking at the attached driver project, I’ve attached the old and new driver LV I was previously running, version Most 5332a Software Boards: I finally found the manual and tried some of the exa Net code for performing frequency measurements using the ?
I got a commands, but commands are supports GPIB port. Where can I reduce frequency resolution.
Actualyl in the default program, which I donwloaded from NI website, there was some unnecessary commands which were making whole devices slowe. Message 1 of 5.
Agilent X Series Counter/Timer – IEEE (GPIB) Driver for LabVIEW – National Instruments
I modified this file by putting a while loop to. I don’t have GPIB.
Is it possible to measure the time between two pulses that are about 1 second e. The examples I’ve found seem to be for an old Looking at the attached driver project, I’ve attached the old and new driver LV These timeout issue appears when the connection between Hello everybody, I have a problem to want to ask everybody. I am trying to set a continuous totalize on channel one which i managed to do pretty easily in vb using: I want to communicate with the A fr counter via RS Hi After update my A from firmware 1.
We have 5 systems all with one of these instruments, and it is the only instrument giving us any problems. Then I’ve tried to install that add-in manuall I need to make successive second 2 hour period measurement and I want them to be gapless.
Agilent 5313X Series Counter/Timer
Log in to follow, share, and participate in this community. General Use Programming and Remote Communications clear selected.
So where can I set resolution to like 1Hz so that I can take more than 2 measurements in a second. I’m trying to make a Labview program for the Agilent NA load box to regulate the voltage and drive a aggilent from the load. PPS signal apart using A??
Message 4 of 5.